|Home||About us||Commercial Photography||Print Gallery||Articles/reviews/blog|
Spyder 3 Pro - Review
Monitor calibration, with ambient light measurement
The Spyder 3 range is an all new profiling and calibration package from Datacolor.
This review looks at the Spyder 3 Pro version which includes dual monitor support.
We've looked at it on an Apple Mac system, but it works just the same on a PC (inc. vista)
A lot of people ask me 'Why should I bother about calibrating my monitor?'
Well, if knowing whether what you see on your screen is an accurate version of the colours in an image is important, then it's a vital first step in getting better results, whether printing your own work, or sending it to others.
The Spyder 3 range features a new design of sensor, used to measure light from your display (LCD/laptop/CRT).
The key features of the Spyder 3 Pro are listed as: (info from Datacolor)
The sensor is a new design (compared to the old Spyder 2) that comes with a stand.
The underside view shows the sensor (behind the holes) that measures the light that comes off your screen, in order to carry out the calibration and profiling process.
You can also see the (removable) suction cap, one means of holding the sensor in place during measurements.
The new Spyder 3 sensor is larger (27mm diam.) than the Spyder 2 to get better and more repeatable readings from screens.
The light measuring area has gone from 165 mm2 to 372 mm2, while the measurement accuracy is stated as going from 0.0035 (x,y) in the Spyder2 to 0.0025 in the Spyder3 (figures from Datacolor).
Calibration and times have also fallen, and there is now a new 'recalibrate ' option.
I don't like giving spurious numbers for timings, which depend rather a lot on the particular computer you are using, but my feeling is that calibration times (on my Macs) are about two thirds of the Spyder 2, and the recalibrate option is about half the full calibration time.
If you're curious as to what's behind the sensor aperture, then this cut-away shows the seven coloured filters used.
A reading is taken through each one to decide what colour your screen is showing.
The software is installed from the CD. You get two different applications, the Spyder3Utility and the Spyder3Pro application itself.
The Utility is intended to run at start-up and enables realtime ambient light monitoring and access to the main application. It also offers an optional confirmation that all the elements of your calibration are in place and set up correctly. The optional checks are particularly useful for PC users where it is easier to have different aspects of your system's colour management messed up by other applications.
These are the things checked (if you want ... the default is off)
Another program from Datacolor with help that is actually helpful...
I tried the Spyder 3 Pro out on my Mac PowerBook and G5 Desktop / Apple 23" LCD. Using the Spyder 3 Pro on a PC windows system looks very similar.
To start with the software checks what monitors you have connected. In this case just the one.
It's spotted the old monitor profile I was using at the time and offers me the choice between checking that the existing profile is OK (a shorter process) or recalibrating.
At 48 days old, it's time to recalibrate (I normally do this every 3-4 weeks)
First, decide what sort of display you are using.
The projector option doesn't work for the Spyder 3 Pro (I've discussed it in the Spyder3Elite review)
I've selected LCD.
Next some information is required about the particular controls your monitor has (if any)
Also the colour controls (if any)
With not many controls on my monitor, I just get to check the luminance.
Note that with the Spyder3Pro, you do not get the option to set an absolute level. This is perfectly good enough for many users, but if you need absolute settings then consider the Spyder3Elite.
As ever with Datacolor software, there is genuinely helpful 'help' available. It opens up in a web browser and is comprehensive enough to cover most issues.
You can now choose the settings for your calibration.
I've shown all of them here, but you could just pick the default, or use the ambient light measurement feature via the SpyderUtility to suggest a setting for your particular room lighting settings.
Non grey balanced calibration may suit some types of display (not common so it defaults to on)
It's at this point you can use the ambient light detection feature suggest a setting (this defaults to off)
Your settings are displayed, to confirm, before proceeding
To make measurements, the sensor needs to be in contact with the screen.
You can either use the suction cup (good for CRTs) or remove it.
There is also the small weight on the lead to counterbalance the sensor if you don't use the suction cup.
The weight on my sample was a little tight on the thin cable, so do be careful if you need to move it.
Although it is suggested that using the suction cup with LCDs is OK, my own preference is to use gravity, by just tilting the screen back a bit and removing the suction cup.
The dialogue below appears when you first use the software (you can change it later, via the preferences)
I've chosen not to stick the sensor to my screen. This means the sensor is shown the correct way up on the calibrations screens.
Place your sensor against the screen and start the measurement process.
A series of coloured patches are shown and the sensor measures the light coming from your screen
After a few minutes you get to save your profile
After this you can use the 'SpyderProof' page to see just how much difference the profile has made to your display.
You can zoom in to any of these images and see the differences.
That's it - your monitor is now calibrated.
If you have multiple monitors then you can go back and recalibrate the next one.
One other thing to remember - many PC laptops may support an external screen, but only allow you to have one active profile working at a time, so you can have -either- the laptop screen or the projector calibrated at any one time.
For most users, a simple run through the calibration process is all that's needed.
One useful feature is being able to do a quick check of the validity of an existing profile, by the 'CheckCAL' process right at the start.
Here I'm looking at a profile only made an hour or so previously.
A shortened calibration cycle gives you a check on whether the profile is still relevant
Don't be too picky when checking those numbers, some slight variation is to be expected.
The software preferences allow you to choose the type of ICC profile produced
and to decide how often (if at all) the software will remind you about profile age
All quite straightforward and, as ever, with well written help information easily at hand.
I've tried the Spyder 3 Pro on several screens and it gave good consistent results.
It's faster than the Spyder 2 and very easy to use. The defaults are well chosen and meet the needs of the majority of potential users.
The Software is easy to use with comprehensive help information, and should give no difficulties for those wanting basic calibration of monitors.
I've collected together some information from Datacolor that should help compare their current products.
First, the new sensor...
Comparison of S2 vs S3 (figures from Datacolor)
There are quite a few changes in the sensor over the old Spyder2 (still perfectly good by the way)
For a comparison of all the various Spyder options (as of May 2008), there is this info from Datacolor.
* Note that the Spyder3Print (full review) is the new version of the PrintFIX PRO. We have a full review of the PrintFIX Pro. If you have a PrintFIX PRO, then do note that the free software update for the Spyder3Print package will work with your existing spectrocolorimeter.
Spyder products feature comparisons
Information from Datacolor:
Spyder3 V3 vs V4 software features (May 2010) - S3Elite V4 review
Product features (late 2009)
The Spyder 3 Pro has most of the functionality many users would need for calibrating monitors.
Buying the Spyder3 Pro
We make a specific point of not selling hardware, but if you found the review of help please consider buying the Spyder3 Pro, or any other items at all, via our links with Amazon or B&H.
It won't cost any more (nor less we're afraid) but will contribute towards the running costs of our site.
Amazon.de Spyder 3.0 Pro
Amazon.fr Spyder 3 Pro
I have a Sony projector that I use for presentations and lectures, so the 'Elite' functionality is very useful. If I didn't need that, then the 'Pro' package would fulfil my everyday working needs.
One other area that the Elite makes a difference is if you need to calibrate several different systems to be as similar as possible.
Monitor calibrator with support for multiple monitor systems.
Manufacturer details: Datacolor
Price (from Datacolor) $279 (Inc. 2yr. warranty)
Note - If you own an older Spyder there is (until Jan. 20th 2008) a discount upgrade progamme - details at the ColorVision web site
Declaration of interest - Keith was asked to look at beta versions of the software and hardware before its final release, but Northlight Images has no commercial relationship with ColorVision.
See our review policy for more informationBefore calibrating your monitor on Windows PC systems, you should check to see that Adobe gamma is turned off if it was installed. We've got a short guide to removing Adobe Gamma that might be of help.
The views in this article represent those of Keith Cooper.
Keith is always happy to discuss matters raised in his articles. You can Email Us
Northlight Images prides itself on its independence when giving advice. We do not sell hardware or software and have no direct commercial links with any of the software or hardware vendors that may be mentioned here. See our Review Policy for more information.
You can search all the many hundreds of articles and reviews on the site for more information
Have you found an article on the site useful or helpful?
If so, please consider sharing a link to the article or mentioning it on a forum or blog - Thanks to everyone who's helped the site become better known.
Explore our site... Digital Black and White photography and printing - some of Keiths thoughts, techniques and tips for those interested in a digital approach to black and white. There are many hundreds of entirely free articles and reviews on the site. New site content appears on the News, articles and reviews page.Thanks to the visitors who've made Amazon purchases (any kinds of items whatosever)
via: Amazon UK/Amazon France/Amazon Germany/Amazon USA/Amazon Canada
It won't save extra money we're afraid, but it does help in the running of the site, and we really appreciate it...
Commercial Photography - UK Architectural Photography services - Industrial - Print Gallery - Landscape Photographer